Access the full text.
Sign up today, get DeepDyve free for 14 days.
E. Wu, Yin Wu, J. Nicholls, Jie Wang, S. Liao, Shuguang Zhu, C. Lau, H. Tse (2007)
MR diffusion tensor imaging study of postinfarct myocardium structural remodeling in a porcine modelMagnetic Resonance in Medicine, 58
(2011)
Translational dynamics and magnetic resonance
G. Greene, B. Zappone, X. Banquy, Dong Lee, O. Söderman, D. Topgaard, J. Israelachvili (2012)
Hyaluronic acid–collagen network interactions during the dynamic compression and recovery of cartilageSoft Matter, 8
Derek Jones, T. Knösche, R. Turner (2013)
White matter integrity, fiber count, and other fallacies: The do's and don'ts of diffusion MRINeuroImage, 73
P. Basser, J. Mattiello, D. LeBihan (1994)
Estimation of the effective self-diffusion tensor from the NMR spin echo.Journal of magnetic resonance. Series B, 103 3
A. Hellerbach, Verena Schuster, A. Jansen, J. Sommer (2013)
MRI Phantoms – Are There Alternatives to Agar?PLoS ONE, 8
E. Rössler, C. Mattea, S. Stapf (2015)
NMR dispersion investigations of enzymatically degraded bovine articular cartilageMagnetic Resonance in Medicine, 73
(2008)
Multiparametric characterization of healthy and diseased articular cartilage at 17.6T: Early results
D. Pierce, Werner Trobin, J. Raya, S. Trattnig, H. Bischof, C. Glaser, G. Holzapfel (2010)
DT-MRI Based Computation of Collagen Fiber Deformation in Human Articular Cartilage: A Feasibility StudyAnnals of Biomedical Engineering, 38
M. Kubicki, C. Westin, R. McCarley, M. Shenton (2005)
The Application of DTI to Investigate White Matter Abnormalities in SchizophreniaAnnals of the New York Academy of Sciences, 1064
C. Pierpaoli, P. Basser (1996)
Toward a quantitative assessment of diffusion anisotropyMagnetic Resonance in Medicine, 36
Philip Batchelor, David Atkinson, Derek Hill, Derek Hill, F. Calamante, Alan Connelly (2003)
Anisotropic noise propagation in diffusion tensor MRI sampling schemesMagnetic Resonance in Medicine, 49
W. Press, S. Teukolsky, W. Vettering, B. Flannery (2003)
Numerical Recipes in C++: The Art of Scientific Computing (2nd edn)1 Numerical Recipes Example Book (C++) (2nd edn)2 Numerical Recipes Multi-Language Code CD ROM with LINUX or UNIX Single-Screen License Revised Version3European Journal of Physics, 24
Y. Assaf, O. Pasternak (2007)
Diffusion Tensor Imaging (DTI)-based White Matter Mapping in Brain Research: A ReviewJournal of Molecular Neuroscience, 34
Ross Crawford, Sally Visser, James Pope (2008)
Structural adaptations in compressed articular cartilage measured by diffusion tensor imaging.Osteoarthritis and cartilage, 16 1
X. Deng, M. Farley, M. Nieminen, M. Gray, D. Burstein (2007)
Diffusion tensor imaging of native and degenerated human articular cartilage.Magnetic resonance imaging, 25 2
E. Farrher, J. Kaffanke, A. Celik, T. Stöcker, F. Grinberg, N. Shah (2012)
Novel multisection design of anisotropic diffusion phantoms.Magnetic resonance imaging, 30 4
S. Skare, M. Hedehus, M. Moseley, Tie-Qiang Li (2000)
Condition number as a measure of noise performance of diffusion tensor data acquisition schemes with MRI.Journal of magnetic resonance, 147 2
M. Seiler, F. Seiler (1989)
Numerical Recipes in C: The Art of Scientific ComputingRisk Analysis, 9
P. Kuchel, G. Pagés, K. Nagashima, S. Velan, Vimalan Vijayaragavan, V. Nagarajan, Kai Chuang (2012)
Stejskal–tanner equation derived in full†Concepts in Magnetic Resonance Part A, 40
S. Roosendaal, J. Geurts, H. Vrenken, H. Hulst, K. Cover, J. Castelijns, P. Pouwels, F. Barkhof (2009)
Regional DTI differences in multiple sclerosis patientsNeuroImage, 44
Mark Bastin, Paul Armitage, Ian Marshall (1998)
A theoretical study of the effect of experimental noise on the measurement of anisotropy in diffusion imaging.Magnetic resonance imaging, 16 7
E. Altschuler, T. Williams, E. Ratner, R. Tipton, R. Stong, F. Dowla, F. Wooten (1997)
Possible Global Minimum Lattice Configurations for Thomson`s Problem of Charges on a SpherePhysical Review Letters, 78
(2005)
Highresolution diffusion tensor imaging of human patellar cartilage: Feasibility and preliminary findings
L. Filidoro, Olaf Dietrich, J. Weber, E. Rauch, T. Oerther, M. Wick, Maximilian Reiser, Christian Glaser (2005)
High‐resolution diffusion tensor imaging of human patellar cartilage: Feasibility and preliminary findingsMagnetic Resonance in Medicine, 53
P. Basser, Derek Jones (2002)
Diffusion‐tensor MRI: theory, experimental design and data analysis – a technical reviewNMR in Biomedicine, 15
L. Axel, V. Wedeen, D. Ennis (2014)
Probing dynamic myocardial microstructure with cardiac magnetic resonance diffusion tensor imagingJournal of Cardiovascular Magnetic Resonance, 16
Radiology and Oncology Ljubljana Slovenia www.radioloncol.com research article Franci Bajd1,2, Carlos Mattea1, Siegfried Stapf1, Igor Sersa2 TU Ilmenau, Institute of Physics, Fachgebiet Technische Physik II, Ilmenau, Germany Jozef Stefan Institute, Ljubljana, Slovenia Radiol Oncol 2016; 50(2): 175-187. Received 27 August 2015 Accepted 8 February 2016 Correspondence to: Igor Sersa, Ph.D., Jozef Stefan Institute, Jamova 39, SI-1000 Ljubljana, Slovenia. Fax: + 386 1 477 3191; E-mail: igor.sersa@ijs.si Disclosure: No potential conflicts of interest were disclosed. Background. Diffusion tensor imaging exploits preferential diffusional motion of water molecules residing within tissue compartments for assessment of tissue structural anisotropy. However, instrumentation and post-processing errors play an important role in determination of diffusion tensor elements. In the study, several experimental factors affecting accuracy of diffusion tensor determination were analyzed. Materials and methods. Effects of signal-to-noise ratio and configuration of the applied diffusion-sensitizing gradients on fractional anisotropy bias were analyzed by means of numerical simulations. In addition, diffusion tensor magnetic resonance microscopy experiments were performed on a tap water phantom and bovine articular cartilageon-bone samples to verify the simulation results. Results. In both, the simulations and the experiments, the multivariate linear regression of the diffusion-tensor analysis yielded overestimated fractional anisotropy with low SNRs and with low numbers of applied diffusion-sensitizing gradients. Conclusions. An increase of the apparent fractional anisotropy due to unfavorable experimental conditions can be overcome by applying a larger number of diffusion sensitizing gradients with small values of the condition number of the transformation matrix. This is in particular relevant in magnetic resonance microscopy, where imaging gradients are high and the signal-to-noise ratio is low. Key words: microscopy; diffusion tensor imaging; anisotropy; signal-to-noise ratio; cartilage Introduction Diffusion tensor imaging (DTI) is a widely used magnetic-resonance imaging (MRI) technique, which enables noninvasive assessment of structural integrity of fibrous tissues with a high degree of anisotropy, such as brain white matter and myocardium.1,2 Specifically, the technique could be exploited for a dynamical follow-up of minor anisotropy alternations due to tissue structural changes arising during progressive disease development, such as schizophrenia3, multiple sclerosis4 and myocardium infarct.5 The method is gaining clinical interest also in applications to tissues with less expressive anisotropy or highly localized compartments with increased level of fiber alignment, Radiol Oncol 2016; 50(2): 175-187. such as articular cartilage6,7, which is a relatively thin tissue with a thickness of up to few millimeters and has a depth-dependent collagen fiber architecture. In DTI, the basic assumption is that diffusive motion of spin bearing particles within the tissue is determined by an alignment of tissue fibers; hence their diffusional anisotropy directly corresponds to anisotropy of the restrictive fibers. The method basically consists of an imaging part, usually employing spin-echo based MRI pulse sequences8, to which a pair of diffusion sensitizing gradients (DSG) is added in order to encode magnetic resonance (MR) signal of spin bearing particles with diffusive motion, resulting into a diffusion-attenuated MR signal. In order to obtain sufficient information on anisotropy of diffusive doi:10.1515/raon-2016-0018 Bajd F et al. / Signal-to-noise ratio in diffusion tensor MR microscopy motion, DSG must be applied in at least six noncoplanar directions to determine six independent elements of the laboratory-frame diffusion tensor.9 In the DT-MRI analysis, diffusion anisotropy is calculated by transforming the laboratory-frame diffusion tensor into the principal frame of reference using diagonalization.10 Determination of diffusion anisotropy can be biased due to instrumentation imperfections9, such as non-optimally calibrated DSG, and due to postprocessing errors.11,12 It was shown, that a number and directionality of the applied DSG configuration play an important role in a noise propagation in DTI post-processing analysis.12 Specifically, noise propagation in DTI, resulting to noise-induced rotational variance of diffusion tensor ellipsoid, can be reduced by decreasing a condition number ( ) of the b-matrix12 as well as by increasing the ).11 Therefore, attempts signal-to-noise ratio ( were made to find a robust measure for diffusion anisotropy, such as the lattice index.13 Among all the proposed measures fractional anisotropy (FA) became commonly accepted. Accuracy in determination of diffusion tensor is of a great importance in biomedical imaging as falsely measured tissue anisotropy could lead to clinical misinterpretations and inappropriate treatment decisions.14 Reliability of the DTI method can be efficiently tested either by using perfectly aligned fiber phantoms with an a priori known anisotropy, yielding anisotropic diffusion along the preferential fiber orientation15,16, or by using completely isotropic materials. In both cases, overestimated apparent anisotropy could arise as an undesirable consequence of the DTI analysis. A fundamental quesand a tion is, how the DTI factors, specifically choice of a DSG configuration, influence accuracy of a diffusion tensor determination. This issue is specifically important in diffusion tensor magnetic resonance microscopy (DT-MRM), in which is usually low due a high diffusion weighting and due to a small voxel size, respectively. The effect of low SNR is more pronounced in some biomaterials with anisotropic diffusion that exhibit short T2 relaxation time, as for example articular cartilage.17,18 The main motivation for this study is analysis of factors influencing diffusion anisotropy in DTMRM signal post-processing. The study is organized into two parts. In the first part, the effect of noise propagation from synthetic raw DTI data to the diffusion tensor eigenvalues is examined theoretically for different DSG configurations: selected commonly used, random and isotropic DSG configurations. In the second part, the theoretical Radiol Oncol 2016; 50(2): 175-187. results are verified experimentally. DT-MRM was performed for two different materials, tap water with isotropic diffusion and bovine articular cartilage-on-bone samples before and after compression. The study is in particular focused to unfavorable experimental conditions that often arise in DT-MRM and could result in biased diffusional anisotropy. Materials and methods Theoretical background and simulations Single-voxel DTI data of an isotropic medium with were generated as a diffusion constant equal to -dimensional column vector ( being the a number of DSG directions) containing normalized magnitude MR signal intensities [1] where fusion attenuation is referred to as difand is gyromagnetic ratio, is a noise factor. Here, is DSG amplitude, is duration of an individual DSG pulse, is time-separation between the two is signal-to-noise ratio in the DSG pulses and pulsed field gradient (PFG) pulse sequence. The noise was introduced to the generated DTI data by -dimensional noise vector , of which coma are uniformly distributed ranponents dom numbers, which are included in the noise factor. Seventeen commonly used DSG directions12, presented in Table 1, as well as different random and isotropic DSG directions with were considered. With random DSG configurations, directions of diffusion sensitizing gradients were modeled as [2] where , and are three evenly distributed random numbers with . Isotropic DSG configurations were obtained from the corresponding random ) by numerically DSG configurations (for each minimizing their average Coulomb-like interaction energy (i.e., the Thompson's problem20) , [3] Bajd F et al. / Signal-to-noise ratio in diffusion tensor MR microscopy using Monte Carlo simulation approach.21 For each DSG, its transformation matrix was calculated using definition [4] where [5] and the effective gradient for a spin echo-like DTI pulse sequence including a pair of DSG gradients is equal to [6] The 3 x 3 b-value matrix defined in Eq. 4 has only six different elements that can be arranged into a 1 x 6 raw vector with elements . The raw vecx 6 DTI tors can be further arranged into a transformation matrix . A condition number of . The the matrix was calculated as components of the diffusion tensor in the laboratory , frame of reference, were calculated by solving an over-determined , in a form of system of equations, [7] The laboratory-frame diffusion tensor, TABLE 1. A list of the analyzed commonly used diffusion sensitizing gradients (DSG) configurations in DTI, adopted from12, along with the corresponding values of , and . # 1 2 3 4 5 6 7 8 9 10 11 12 13 14 15 16 17 Scheme name Tetrahedral Cond 6 Decahedral Jones noniso Dual-gradient Jones 10 Jones 20 Jones 30 Papadakis Jones 6 Muthupallai Tetraortho DSM 6 DSM 10 DSM 20 DSM 30 DSM 40 6 6 10 7 6 10 20 30 12 6 6 7 6 10 20 30 40 9.148 5.984 2.748 2.560 2.000 1.624 1.615 1.594 1.587 1.583 1.581 1.527 1.323 1.324 1.668 1.430 1.401 16.53 14.19 10.70 12.13 11.44 9.67 8.10 7.16 9.29 11.04 11.11 10.68 11.41 10.02 8.43 7.45 6.87 CN = ondition number [8] and were calculated as a the corresponding ( ) and ( ) function of different noise vectors mimicking for different experiments of which results were then averaged in order to reduce their randomness: [11] was then diagonalized to the principal-axis-frame diffusion tensor [9] The diffusion tensor eigenvalues , and were used to calculate the average diffusion coeffi, and cient, the fractional anisotropy, defined as10 [10] For each DSG configuration, either taken from the Table 112 or calculated as random or isotropic directions, diffusion tensor eigenvalues as well as The diffusion tensor quantities were then averaged over the characteristic window of the domain ( ) to obtain their representative characteristic scalar valand . For random ues denoted as Radiol Oncol 2016; 50(2): 175-187. Bajd F et al. / Signal-to-noise ratio in diffusion tensor MR microscopy FIGURE 1. A simulation flowchart for analysis of selected commonly used and random/isotropic diffusion sensitizing gradients (DSG) configurations. DSG configurations, a relation between was modeled with a power-law function and [12] where and are two fitting parameters. In the simulations, water-like isotropic medium with was considered. Simulations were performed using an in-house written program, developed within the Matlab programming environment (MathWorks Inc., Natick, MA, USA). Flowcharts presenting the simulation algorithm for the simulations including DSG configurations from Table 1 and random or isotropic DSG configurations are shown in Figure 1. Additional numerical simulations were performed in order to investigate the effect of noise propagation in tissues with low diffusional anisotropy. In these simulations, diffusion tensor in Radiol Oncol 2016; 50(2): 175-187. the principal frame of reference was considered as a prolate spheroid, which was, for the sake of convenience, oriented with the largest dimension along the z-axis of the laboratory frame of reference (zLAB). In this direction, non-restricted diffusion was assumed, i.e., the corresponding eigenvalue was equal to D0, while diffusion was reduced to D0 (0 1) in the other two orthogonal directions. The principal-frame diffusion tensor was hence modeled as [13] The primary eigenvector (corresponding to , the largest eigenvalue) was equal to while the other two eigenvectors were chosen as and . The eigenvec- Bajd F et al. / Signal-to-noise ratio in diffusion tensor MR microscopy tors determined the change-of-basis transformation matrix between the principal and the laboratory frame of reference: [14] dimension. As the read gradients in 1D DT-MRM were applied only during or close to MR signal acquisition, their contribution to the transformation matrix was minor and was therefore neglected. Prior to the experiments, Stejskal-Tanner plots were measured for each gradient channel. The gradient channels were then calibrated to yield a diffusion constant of water at room temperature. The transformation of the diffusion tensor into the laboratory frame of reference was calculated . In the case of the aboveas defined prolate spheroid, components of the diffusion tensor in the laboratory frame of reference were expressed by where . Normally distributed (Gaussian) noise was added to the MR signal intensities. For a given preset FA value and for a given DSG configuration, the results of simulations are presented by a difference between the calculated fractional anisotropy FA' and the preset fractional anisotropy . In addition, an orientation difFA, i.e., by ference between the preset and the calculated pri, is premary eigenvector, i.e., sented as well. DT-MRM of articular cartilage Bovine cartilage-on-bone samples, containing an intact cartilage tissue and the underlying part of a subchondral bone, were carefully dissected from fresh stifle joints of bovine femur bones (provided by a local meat provider) using a commercially available bow saw and a dentist driller set (Meisinger, Neuss, Germany). Samples were cut into cylindrically shaped pieces with a diameter of 6 mm and with an average height of 8 mm, fitting to an NMR tube with an inner diameter of 7 mm. After dissection, the samples were washed in physiological phosphate buffer saline (PBS) and sealed into plastic bags for deep-freezing storage.22 Prior to DT-MRM experiments, each sample was allowed to spontaneously defreeze at temperature of 8°C. Then, the sample was inserted into an NMR tube and immersed into Fluorinert FC-70 (SigmaAldrich, Germany), which was used to prevent samples from desiccation. Compression of articular cartilage was obtained by loading a plastic indenter, positioned above the articular surface, with at the inweight-induced pressure of denter-cartilage interface. After the application of an external pressure, each sample was allowed for spontaneous equilibration for 2 hours.17 DT-MRM experiments on articular cartilage-onbone samples were performed on a 7-Tesla vertical-bore superconducting magnet equipped with microimaging accessories and controlled by the Avance spectrometer (Bruker, Ettlingen, Germany). The gradient system had maximum gradients of 1 T/m and slew rate of 4000 mT/m/ms. Due to relatively fast transversal relaxation processes in cartilage tissue, DT-MRM was performed using a stimulated-echo pulse sequence. The following imaging parameters were used: imaging matrix 256 x 128 (or 128 x 64), field of view 20 x 10 mm, isotropic in-plane resolution of 78 um (or 156 um), slice thickness 2 mm, dwell time 10 s (or 20 s), number of averages . 32, echo and repetition time and A DSG configuration with was used.18 Square-shaped DSG pulses with and were used, Radiol Oncol 2016; 50(2): 175-187. DT-MRM of tap water DT-MRM experiments of a tap water phantom with a cone shape were performed on a horizontal-bore 2.35-T MRI scanner (Oxford Instruments, Abingdon, United Kingdom), equipped with microimaging accessories (Bruker, Ettlingen, Germany) and controlled by a Tecmag Apollo spectrometer (Tecmag, Houston TX, USA). The gradient system had top gradients of 0.25 T/m and slew rate of 1200 mT/m/ms. For acquisition of onedimensional DTI profiles along cone axis, a spinecho based 1D DT-MRM sequence was employed. For the sequence, the following imaging parameters were used: 256 acquisition points, field of view 40 mm, spatial resolution of 156 m, dwell time 10 s, number of averages 4 (with the half-Cyclops phase cycling scheme), echo and repetition . Square-shaped DSG time and pulses with were used, yielding the corresponding diffusion . 1D DT-MRM attenuation equal to was examined for isotropic DSG configurations . The contribution of imaging with gradients to the DTI transformation matrix was minimized using non-selective (hard) excitation RF pulses and by performing imaging in just one Bajd F et al. / Signal-to-noise ratio in diffusion tensor MR microscopy FIGURE 2. Simulated two dimensional maps of , and fractional anisotropy four representative, commonly used diffusion sensitizing gradients (DSG) configurations from Table 1 (#1, #8, #13 and #16). for yielding the corresponding diffusion attenuation . Again, the imaging gradients equal to (read-dephase and phase gradients) were applied immediately before MR signal acquisition in order to minimize their contribution to the transformation matrix . All gradients, the imaging gradients and DSG, were considered in the numerical calculation of the transformation matrix elements (according to Radiol Oncol 2016; 50(2): 175-187. Eq. 4). All DT-MRM analyses were performed within the Matlab programming environment. Results Simulated 2D maps over the the second largest eigenvalue domain of , of the Bajd F et al. / Signal-to-noise ratio in diffusion tensor MR microscopy difference among the smallest and the largest eigenvalue and of the corresponding fractional anisotropy for four representative DSG configurations, i.e., the DSG configurations #1, #8, #13 and #16 from Table 1, are presented in Figure 2. Two of the selected DSG configurations correspond to the commonly used and #8 with DSG configurations (#1 with in Figure 3A,B, respectively), while the other two correspond to the DSG configurations, optimized in12 with respect to the condition num(again with and ). It can ber be seen that the largest deviations of the studied diffusion tensor quantities from their ideal values from 0 and from 0) are found ( from , and low values of . From at high values of the result in Table 1, it is evident that region proportions of poorly determined DT eigenvalues, i.e., dark red regions in maps in Figure 2 defined deby the condition . The results in Table 1 crease with a decreasing also indicate that for all DSG configurations, the proportion could be additionally decreased by . increasing domain-averaged valFigure 3 shows ues of diffusion tensor eigenvalues , and (Figure 3A,B) along with the corresponding aver(Figure 3C,D) and age diffusion coefficient (Figure 3E,F) as a function fractional anisotropy for random DSG configurations with differof . ent The results are shown in two formats, i.e., the calculated quantities in a log format with a broad are shown range of condition numbers in Figure 3A,C,E, while the same quantities in a lin(zoomed ear format with the range of gray regions of the corresponding log graphs) are shown in Figure 3B,D,F. From the graphs in Figure 4 it can be seen that random DSG configuraand high overestimate the tions with low largest diffusion tensor eigenvalue and underestimate the smallest eigenvalue, while the remainand ing eigenvalue remains independent on holds for a broad range of . the equality remains non-zero Therefore, the difference ) also for the smallest (of the order of values. Interestingly, as the difference is symmetric with respect to , its undesired contribution is , of which valcanceled in the calculation of ues are hence only weakly scattered around in a . On the contrary, the symmetric broad range of is not canceled in the calculation difference , which thus contributes to an existence of of the apparent fractional anisotropy of a generic iso- FIGURE 3. Panels on the left show simulated average values of diffusion tensor (C) and of fractional eigenvalues , and (A), of average diffusion constant (E) as a function of condition number for random diffusion anisotropy sensitizing gradients (DSG) configurations. Panels on the right (B,D,F) display the . Solid-line zoomed left-side panels for condition numbers in the range curves (E,F) correspond to best fits of the model function [Eq. 12] to the simulated data. The graphical insert in panel (A) illustrates the selected random DSG . configuration with in the range of , tropic medium. With span range between 0.01 and 0.08. the values of was obtained In Figure 4, the minimal . Clearly, apparent fractional anisotwith ropy is highly dependent on the condition number. Solid curves in Figure 3E,F represent best fits of the vs. model function in Eq. 12 to the simulated data calculated for each independently. As can be seen from the graphs in Figure 3E,F, the , while for model fits well to the data for the model is less accurate due to satura. tion of Figure 4 shows maximal, average and minimal (Figure 4A) as well as the optimally values of and , along fitting parameters of Eq. 12, i.e., Radiol Oncol 2016; 50(2): 175-187. Bajd F et al. / Signal-to-noise ratio in diffusion tensor MR microscopy FIGURE 4. Characteristic values of the simulated correlations between and for random diffusion sensitizing gradients (DSG) configurations : maximal, average and minimal values of (A), best fit as a function of , and fit quality (B). The arrow designates a crossover, at which parameters . The apparent fractional anisotropy drops below fractional anisotropy graphical insert in panel A illustrates the selected random DSG configuration with . FIGURE 5. Simulated average fractional anisotropy (solid (void symbols) and the corresponding condition number for random (blue symbols) and symbols) as a function of isotropic (red symbols) diffusion sensitizing gradients (DSG) configurations. The graphical insert illustrates a distribution of DSG directions in a random and an isotropic DSG configuration . with with the corresponding values of as a function for random DSG directions (Figure 4B). The of data shown in Figure 4 are taken from Figure 3E,F. From Figure 4A it can be seen that the maximal and as well as the difference between them minimal . At , decrease with an increasing drop to apboth, the maximal and minimal proximately 0.01. In addition, as can be seen from Figure 4B, best fit parameters are monotonically . However, fit quality dedecreasing for from at to creases with larger at . (solid symAverage fractional anisotropy bols) and the corresponding condition number Radiol Oncol 2016; 50(2): 175-187. (void symbols) as a function of , for both random (blue symbols) and the corresponding isotropic (red symbols) DSG configurations, are shown in Figure 5. The data shown in Figure 5 were obtained by averaging the corresponding quantities that were calculated with two different random seeds. As can be clearly seen from the blue curves, the condition number is for random DSG with a configurations in the range of plateau value of , while average fractional anisotropy is a monotonically decreasing function of . Both and average fractional anisotropy and less values are more scattered at smaller . With isotropic DSG conscattered at larger figurations (red curves), however, both average fractional anisotropy and condition number at a are comparatively smaller than with given the corresponding random DSG configurations. Moreover, the condition number decreases with an to approximately , where increasing , it attains a constant plateau value of while the average fractional anisotropy monotoni. A significally decreases with an increasing cant drop of the average fractional anisotropy from at to is noticed for both isotropic as well as at random DSG configurations. Figure 6 depicts results of numerical simulations that were performed with three different preset fractional anisotropies (FA = 0.0, 0.1, 0.3) and with two different SNRs (SNR = 5, 30). In the simulations random and isotropic DSG configurations with NDSG = 6-100 were considered, while diffusion attenuation was held constant by setting G0 = 0.3 T/m and / = 2/15 ms. For each DSG configuration, the results are displayed as an average of N = 100 simulation runs of different noise vectors. As and the curves can be seen, the curves of are decreasing with an increasing NDSG and of SNR; both curve types are decreasing also with an increasing preset FA. Interestingly, the difference is largest with small NDSG 10 with both of random and isotropic DSG configurations (irrespective of the preset FA value and SNR). Experimental results of a tap water phantom, examined by 1D DT-MRM are shown in Figure 7 with stack plots of 1D profiles of diffusion tensor , , , and ) as a function of quantities ( . The 1D profiles were measured with isotropic DSG configurations (the same configurations as in Figure 6) along the axis of the cone-shaped phantom. As can be seen from the 2D MR image (without slice selection) of the phantom and the corresponding 1D intensity profile along the phantom Bajd F et al. / Signal-to-noise ratio in diffusion tensor MR microscopy FIGURE 6. The results of numerical simulations with three different fractional anisotropy (FA) values: 0, 0.1, and 0.3 for isotropic (FA = 0) vs. anisotropic (FA = 0.1, 0.3) case. axis, SNR decreases monotonically towards the tip of the cone-shaped phantom. From the stack plots it can be seen that diffusion tensor quantities are . The biased in the phantom regions with poor . This results can be improved by increasing , where can be well seen in the stack plot of ) coris the lowest in the encircled region ( and the central phantom responding to . region with high Figure 8 shows the experimental DT-MRM results of two bovine cartilage-on-bone samples, i.e., and as well as the corresponding maps of , and , obtained before and after its maps of . The maps were calcompression with culated from the corresponding magnitude MR images obtained with an isotropic in-plane resolution of either 78 m (Figure 8A) or 156 m (Figure 8B). In Figure 8 with the compressed sample, three various regions of interest (ROI1-ROI3) are delinewas determined. While ROI1 ated, from which designates the indenter region (providing no MR signal) for background noise determination, ROI2 and ROI3 designate two regions of weakly and fully compressed cartilage, respectively. The corvalues were in the higher-resoluresponding and , tion MR images equal to and while they were equal to in the lower-resolution MR images. Average values Radiol Oncol 2016; 50(2): 175-187. Bajd F et al. / Signal-to-noise ratio in diffusion tensor MR microscopy FIGURE 7. Experimental 1D DT-MRM results of the cone-shaped water phantom: 2D MR image of the phantom with the corresponding , signal-to-noise ratio (SNR) 1D profile along the phantom axis (top left) and stack plots of the diffusion tensor quantities ( , , ) as 1D profiles as a function of . Isotropic diffusion sensitizing gradients (DSG) average diffusion coefficient [ADC] and configurations were used. The white dotted curve depicts the region with noticeably reduced fractional anisotropy. TABLE 2. Average values of average diffusion coefficient (ADC) and fractional anisotropy (FA) in three different regions of an uncompressed and compressed cartilage sample obtained with two different spatial resolutions Higher resolution (78 m) [10-9 m2/s] Uncompressed cartilage Compression zone Liquid droplet 0.99±0.13 0.63±0.42 1.35±0.11 [1] 0.27±0.13 0.87±0.27 0.24±0.06 Lower resolution (156 m) [10-9 m2/s] 1.12±0.14 1.01±0.82 1.34±0.19 [1] 0.14±0.08 0.82±0.23 0.11±0.04 Discussion The aim of this study is to analyze the effect of the signal-to-noise ratio and DSG configuration on noise propagation in the DT-MRM post-processing analysis for the isotropic (FA = 0) as well as for anisotropic case (FA > 0). The principal findings of the study are: i) noise propagation in the DT-MRM analysis is manifested in an increased deviation of diffusion tensor eigenvalues; ii) deviations of diffusion tensor eigenvalues result to an overestimation of fractional anisotropy, while the average diffusion coefficient remains unchanged; iii) fractional anisotand ropy bias could be reduced by increasing by optimizing a DSG configuration to a small condition number at a large number of DSG directions. The analysis of numerical simulations is based on correlating the diffusion tensor quantities of ADC = average diffusion coefficient; FA = fractional anisotropy of and in three different regions (uncompressed cartilage, compression zone, liquid expelled from the cartilage tissue) of uncompressed and compressed cartilage obtained with two different spatial resolutions are given in Table 2. Radiol Oncol 2016; 50(2): 175-187. Bajd F et al. / Signal-to-noise ratio in diffusion tensor MR microscopy FIGURE 8. Experimental DT-MRM results of articular cartilage-on-bone samples in a form of 2D maps of , , , average diffusion coefficient (ADC) and fractional anisotropy (FA). The maps were obtained with a spatial resolution of either 78 m (A) or 156 m (B). Solid white boxes indicate three different ROI regions for signal-to-noise ratio (SNR) determination, while dashed white lines encircle droplets of liquid expelled from the cartilage tissue. White arrows denote the compression zone. isotropic medium with the condition number of the transformation matrix and the number of DSG directions. It was evidently shown (Figures 3-5) that the extent of the fractional anisotropy overestimation is dependent of the both parameters. Interestingly, noise propagation with random DSG configurations appears in a form of a symmetric deviation of the largest and the smallest diffusion tensor eigenvalue from the expected value of (Figure 3B), while the second diffusion tensor eigenvalue remains practically unchanged over the entire range of condition numbers. The deviation Radiol Oncol 2016; 50(2): 175-187. Bajd F et al. / Signal-to-noise ratio in diffusion tensor MR microscopy contributes to an apparent fractional anisotropy considerably, while the deviation is canceled in the calculation of the average diffusion coefficient. With random DSG configurations, an average highly correlates with a fractional anisotropy condition number, which is consistent with et al.12 were obtained with the The minimal values of smallest values of the condition number ( in Figure 3F) corresponding to nearly isotropic DSG configurations, which is in agreement with the results of the study by Batchelor et al.23 This as a is evident from comparison of minimal with random DSG configurations function of as a function of with (Figure 4A) and of isotropic DSG configurations (Figure 5). However, correspond to DSG configurahigh values of tions with one or more preferential directions around which DSG directions are clustered. From the results of simulations, it is also evident that the apparent fractional anisotropy decreases with a large number of DSG directions. Namely, the difference between the maximal and the minimal fractional anisotropy in Figure 4A monotonically decreases with an increasing number of DSG directions. The decrease is also associated with a reduction of the corresponding best fit parameters and the fit quality (Figure 4B). Fractional anisotropy overestimation in isotropic water phantom was studied by DT-MRM only in 1D due to a required large set of isotropic .The experiDSG configurations with different ments were performed in 1D to reduce the total acquisition time and for the phantom cone-shaped water-filled sample was used to obtain continuousalong the phantom axis. In 2D ly decreasing was reduced DT-MRM of articular cartilage, additionally due to a faster transversal relaxation and a reduced voxel size, which resulted into an overestimation of fractional anisotropy. The measured values listed in Table 2 are somewhat larger than those in literature.17 The reported fractional anisotropy values in the uncompressed cartilage in the tangential zone close to are equal to in the intermediate the articular surface, zone with disordered collagen fibers and in the radial zone close to the cartilage-bone interface, while after compression fractional anisotropy and in the compression zone increases to remained unaltered in other zones. A discrepancy between the measured and the reported values can condibe attributed mostly to the unfavorable was the lowest in the compression zone, tions. on account of water redistribution from the zone to the articular surface.24 However, the measured fractional anisotropy values in the subchondral bone region were close to unity, which is in an agreement with another high-field DT-MRM study of articular cartilage.25 In comparison to conventional MRI, in MRM, is usually noise has additional origins. Firstly, low due to a much smaller voxel size. Secondly, in MRM imaging gradients are due to an increased spatial resolution large, which in turn contribute to their interaction with DSG. Interaction between DSG and possible background gradients is possible as well. If the contributions are neither compensated by flipping the signs of DSG on alternate averages9 nor properly considered in the calculation of transformation matrix elements according to Eq. 4, variations of the diffusion attenuated MR signal within an individual voxel could be misinterpreted as a noise which could lead to can an overestimated fractional anisotropy. be improved by optimizing magnetization recovery during each repetition time. The echo time, however, should be set as a compromise between two competing effects, diffusion weighting that increases with the echo time and transversal relaxation that decreases with it. A limitation of the study is that the simulations of anisotropic diffusion were performed only with two different SNR values. However, the selected SNRs were taken from DT-MRM experiment on articular cartilage presented in this study. Another limitation of the study is that the experiments on the water phantom were performed only in one dimension to save experimental time and to thus enable testing of more DSG configurations. The one dimensional DT-MRM approach would be difficult to apply with cartilage samples due to the tissue heterogeneity, which was even more pronounced after the cartilage compression. Conclusions In this study a possible overestimation of fractional anisotropy in DT-MRM was analyzed. It was shown by means of numerical simulations and DTMRM experiments on the isotropic water phantom and low-anisotropy bovine cartilage-on-bone samples that noise propagation from raw data to diffusion tensor eigenvalues can be efficiently reduced by applying DSG configurations with small condition numbers and large numbers of DSG directions. Radiol Oncol 2016; 50(2): 175-187. Bajd F et al. / Signal-to-noise ratio in diffusion tensor MR microscopy
Radiology and Oncology – de Gruyter
Published: Jun 1, 2016
You can share this free article with as many people as you like with the url below! We hope you enjoy this feature!
Read and print from thousands of top scholarly journals.
Already have an account? Log in
Bookmark this article. You can see your Bookmarks on your DeepDyve Library.
To save an article, log in first, or sign up for a DeepDyve account if you don’t already have one.
Copy and paste the desired citation format or use the link below to download a file formatted for EndNote
Access the full text.
Sign up today, get DeepDyve free for 14 days.
All DeepDyve websites use cookies to improve your online experience. They were placed on your computer when you launched this website. You can change your cookie settings through your browser.